HGS911HD Balancer / Vibration Spectrum Analyzer / Data Collector
HGS911HD is an easy operate and small sized economical vibration Spectrum analyzer & data collector. It can be used for overall vibration measurement, spectrum analysis and measurement of high frequency acceleration envelope design makes it very effective for bearing fault detection. HGS911HD testing data can be outputed to computer for database and trending analysis.
It is also a low-cost, easy-to-use balancing meter with the measurement of Amplitude,Rotation speed RPM, & Phase measurement.
Easy to use FFT spectrum analyzer, 400 lines
Measurement of high frequency acceleration envelope makes it a very effective bearing fault detector
Download from computer database point / location ID & upload overall value to database for trending
Powerful standard condition monitoring software
Parameter selection of Acceleration, velocity, displacement and acceleration envelope for FFT.
USB 2.0 interfaces to computer
Date/Time & Battery Status Display
True RMS measurement
RPM, Amplitude, Phase measurement for balancing
Maximum measurement range / Resolution / frequency range (may change):
|Displacement Peak-Peak value||2 mm||1 micron||10 – 500Hz|
|Velocity RMS value||200mm/s||0.1 mm/s||10 – 1000Hz|
|Acceleration Peak value||180m/s2||0.1 m/s2||10 – 15000Hz|
|High Frequency Acceleration Envelope||20unit||0.1unit||5Hz-2kHz #|
# Note: Demodulated from 15kHz-40kHz
Accuracy of vibration measurement:
Noise Level (without vibration input): ACC<0.25 m/s2, VEL<0.5mm/s, Disp<1mm
Non-linearity: ±5% + 2 digits
Frequency response accuracy: ±5% + 2 digits
Rechargeable Li-ion battery, Automatic power off, >20 hours operation time
Operation temperature range: 0 - 50°C
Pickup: Accelerometer with extension stinger and magnetic mount,
Optional tacho-sensor Optional 1 or 2 accelerometers. For 2 accelerometers, a mechanical switch provided.
Rotation speed measurement range: 10-60000 RPM
Storage: 200 sets of ID, acceleration, velocity, displacement, envelope & spectrum
Size: 150 x 70 x 25 (mm)
Weight: 320 gram (including battery)
|The magnetic mount has no attraction force||Take off the protective iron cover on the bottom of the magnetic mount.|
|Cannot power on.||
1. Press and hold the On button for around 5 seconds then release;
2. The voltage of the battery may be low. Recharge the battery.
|Spurious point name appears||
1. Format the memory
2. The battery voltage may be low. Recharge battery and power on again.
|Measurements not stable||
If other faults are found, make detailed records and contact HUATEC Group Corporation, www.huatecgroup.com.
Rechargeable battery is used in HUAEC Group HGS911HDD data collector. It will take several hours to fully charge the battery and then it will be ready for >20 hours of continuous operation.
HUAEC Group HGS911HD and accelerometer should be calibrated periodically. The cycle of calibration is normally one year.
|Contact Person :||JingAn Chen|
|Tel :||8610 82921131,86 13261934319|