350KV Glass tube Penetrant 60mm Directional Radiation Portable X-Ray Flaw Detector XXQ3505
◆ Adopting micro-computer control and modular structure, with strong anti-interference ability and high automaticity and reliable performance.
◆ Parameter setting, working process and trouble indication will be indicated on the LED screen.
◆ Independent aging key is designed for aging the machine conveniently.
◆ Radiating time to cooling time is 1:1.
◆ The controller can recognize the type of x-ray generator automatically.
◆ The water and dust proof color panel and digital sound keyboard are introduced that exposure parameters are promptly and accurately selected.
◆ Ensure safety using the instrument, it is installed with overvoltage, under voltage,
over current, under current and over temperature protection device.
◆ A unique safety lock is designed that protect the operator from the damage of radiation.
◆ Adopting the wider range of KV value, and solve the difficulty problem of short radiographic range.
◆ Exposure delay function ------the delay time can be set less than 10 minutes to keep the operator safe from working place.
◆ Match with various generator and function reliably.
◆ Number of exposure can be accumulated.
(After the unit has been ready, but before press the "ON" key, you can press the "A" key (delay key) three times to check the number of exposure, and press any key to return back.)
◆ With the outlet of ray warning lamp and wire connecting terminal of door-machine interlock.
(Door-machine interlock—when the lead door opens, the instrument cut off the high voltage automatically to protect the operator)
XXQ3505 Glass tube orientation/directional
Main technical parameters
|Max Penetration thickness Steel (A3)||60mm|
|Film Type||Agfa D7|
Double lead foil Intensifying screen
|Input||Voltage (V)||220V±10% 50―60Hz|
|Output||Tube voltage (KVP)||200-350Kv|
|Tube current (mA)||5|
|Operation Mode||Intermittent operation, Radiating time to cooling time is 1:1|
|Circuit Structure||Micro-computer control ,Modular structure,Voice prompt|
|Cooling||Force air cooling|
|Insulation||SF6 Gas Insulation|
This machine is composed of intelligent controller-ray generator, power cable and connecting cable.
1. The controller will consist of the following sections:
Front Panel: LED display, status and failure indicator, touch keyboards (see the introduction of control panel)
Side Panel: power cable socket and connecting cable socket, power switch,key switch, fuse socket, warning lamp and wire connecting terminal of door-machine interlock.
Internal Circuit: power circuit, rectifying circuit, filter circuit, inverter circuit, control circuit and other circuited.
2. The x-ray generator will consist of the following sections:
X-ray tube, High voltage transformer, Temperature relay, Gas pressure gauge and Cooling fan etc.
3. The schematic diagram of the complete machine:
|Directional glass X-ray tube||1 Pcs|
|High voltage joins cable||1 Pcs||25 Meters|
|Power cable||1 Pcs||10 Meters|
|Grounding cable||1 Pcs||5 Meters|
|(Including: alarm lamp,1 Pcs; lead letter, 1 box; Film dark cassettes 5 pcs; intensifying screen 5 pcs; Wire penetrameter (IQI) 1 set; fuse1 pc)|
Testing platform, IQI/wire penetrameter, lead letter/digits, lead tape, Cassette, lead screen, personal dosimeter, film viewer, densitometer, film developer, dark room light, RT calibration block etc.